The Skyline Serices Service Chase will allow the clinical staff to have flexibility and access to overhead electrical, data, medical gas, and other needed medical equipment.
Item number definition
- Pre-installed medical gas outlets
- Pre-installed electrical outlets
- Optional blanks to add additional services.
- Available different lenghts
- 45 degree track system
- Extruded aluminum surface
- All panels are removable
SL-‘’XXX’’ ‘‘XXX’’= Length in inches
Maximum unit length: 100’’
THIS IS A CUSTOM BUILT TO ORDER PRODUCT SO PLEASE CONTACT US AND WE ARE HAPPY HELP FIND THE BEST FIT!
This Ceiling Service Chase unit shall be constructed of .094" thick 6063-t 5 alloy, extruded aluminium for all structural frame members. All exposed aluminium surface shall be clear anodized. Each side corner shall be equipped with Medical Design’s 45º track system as an integral part of the main frame, designed to manage hospital equipment and accessories horizontally.
Bottom faceplate shall be made of stainless steel sheet and shall have pre-punched openings for installation of medical gas outlets, electrical and communication devices. End caps shall be clear anodized and light almond powder coated. All panels shall be removable by means of mounting stainless steel screws located on each side. This product is cULus listed.
Medical gas outlets shall be furnished pre-installed within the bottom faceplate. Connections of all medical gas
outlets are hard piped throughout the length of the unit and terminated with copper tubing located 6’’ above the ceiling level. All components shall comply with National Fire Protection Association (NFPA 99c) & CRN (BNQ 5710-500, CSA Z305.1/B51-95).
Electrical devices shall be supplied and pre-wired by Medical Design. Wiring shall run through metal raceway and terminate at the top of the unit connected to the Hospital electrical system. Conductors shall be No 12 stranded RW-90 copper wire for normal and emergency power circuits. All devices shall be grounded with #10 AWG green jacket stranded copper wire. All electrical receptacles are NEMA 5-15/20R, 125 volts AC, hospital grade and comply with CSA and are cULus listed. Emergency receptacle
Communication devices shall be supplied and installed by others. Provisions for devices shall be coordinated with Medical Design.
All units shall be installed in accordance with Medical Design’s installation instructions. The mechanical and electrical contractor shall provide the final connection to the hospital System. The contractor has the responsibility to insure that every unit is subject to a final site inspection in compliance with all National, State and Local Codes.