Seamless Panel Walls offer you limitless design, increased infection control as well as adaptability for future modifications. The streamlined installation of our panel wall seamlessly integrates all of your services and equipment. This custom designed headwall system features state-of-the-art aesthetics and ensures maximum clinical functionality.
A great fit for:
THIS PRODUCTS IS A CUSTOM BUILT TO ORDER PRODUCT. PLEASE SUBMIT AND RFQ AND WE WILL GET IN-TOUCH WITH YOU ABOUT YOUR PROJECT
- Emergency/Trauma Departments
- Labor and Delivery
- Patient Rooms
- Special Procedure Areas
Materials and Construction
Integrated Accessory Rails (Optional) Numbers of integrated rails (horizontal and vertical) are to match final shop drawings. Accessory rails are clear etched anodized aluminum.
Fascia All fascia panels shall be removable for access and easy installation or maintenance of panel wall services using the spring clip system.
Medical Gas Outlets Amico Corporation, DISS or quick connection type, console style.
Medical Gas Manifold Medical gas distribution shall be supplied by pipe drops to the single point connection above the unit as indicated on the drawing. All medical gas outlets and piping shall be brazed and tested in accordance with CSA Z7396.1-06 standard and NFPA-99c. All medical gas piping shall be Type L copper pipe. Each outlet, piping and manifold shall pass a 24 hour standing pressure test.
Electrical Wiring Wire for standard and critical branch power circuits shall be #10 or #12 (as specified) type THHN stranded copper wire, 600 volt, with heat resistant thermoplastic insulation for hot (black) and neutral (white). Grounds shall be #10 THHN stranded copper wire (green). All ground conductors shall be installed in conduit.
Grounding and Bonding All ground conductors shall be installed in conduit or raceway. Each power receptacle shall have a ground conductor connected to a grounding screw. Where electrical terminations are located inside the Serenity Series Panel Wall, a grounding bus will be installed for each type of power, and shall be installed as to insure grounding for the complete power system.
Low Voltage Data Provisions Shall be connected to device junction box via conduit or raceway. Amico Corporation shall include pull cord extending from junction box to service provision. Telephone nurse call patient station device shall be furnished, installed and wired by low voltage contractor.
Electrical Receptacles Shall be Hospital Grade 15 or 20 amp, 120 or 277 volt, UL listed and marked hospital grade. Amico Corporation shall furnish and install receptacles with wiring required, as shown on submittal shop drawings. Electrical contractor shall insure compatibility of plug on accessory equipment to be used with these devices: Duplex receptacles shall be NEMA style 5-15R or 5-20R, in ivory color for use on normal power circuits, and in red color for use on critical branch power circuits.