The Majestic Series Double Tier Medical Headwall is built for practical use. With multiple horizontal rails, you can easily mount a variety of accessories to suit your needs. This two tier horizontal headwall is also durable and easy to clean. Integrated raceway system allows for future upgrades
- Accessory rails can be used for IT, monitor, cord and equipment management to meet current needs and provide seamless upgrades for future technology
- Rounded top and bottom covers are easy to clean and encourage better infection control
- Injection molded end cap increases durability and makes it easier to clean
- Customizable to meet your facility’s specific needs.
- Expandable to meet your future requirements.
- Powder coated access panels and laminate colors match your room décor
- Services manifolded to a single point for ease of installation
- Available serface mounted or recessed
THIS IS A CUSTOM BUILT-TO-ORDER PRODUCT. PLEASE SUBMIT AN RFQ AND WE WILL GET IN-TOUCH WITH YOU ABOUT YOUR PROJECT!
- Emergency Department/Trauma
- Transitional Step-Down
- Patient Rooms
- Special Procedure
Integrated Accessory Rails
The Double Tier Headwall System has four (4) accessory channels, which are integrated into the aluminum extrusion assembly. The accessory channels are clear etched anodized.
Aluminum strips with High pressure laminate (HPL), color and pattern selection by customer.
Covers and End Caps
Top, middle and bottom cover panels are made of power coated extruded aluminum. End caps are made of injection molded ABS Fire Retardant plastic.
Service Chase (Optional)
Shall be constructed of 6063-T5 extruded aluminum, with passage space for conduit and medical gas piping. High Pressure Laminate MDF inserts shall finish the exposed front, and shall match the laminate color selected.
Medical Gas Outlets
Ohmeda Quick Connect, Chemetron Quick Connect, Puritan Bennett, DISS to match any medical gas outlet type
Direct, Indirect and night light lighting can be provided by the Amico 3' or 4' overbed or ceiling lighting system. These fixtures provide lighting on and above the patient bed.
Materials and Construction
Enclosure shall be constructed of extruded, anodized aluminum alloy sections to provide a modular surface mounted unit with integrated accessory rails for equipment management.
The Majestic system can be manufactured in lengths that range from 12" (305 mm) to 120" (3048 mm). Units that require a length beyond 120" (3048 mm) will be divided into sections using painted connection plates. On-site medical gas and electrical connections between sections will be required.
Each headwall system has integrated raceways and a single area connection for electrical wiring for each type of power (critical, normal and low voltage/communication). All med gas piping shall be hard-piped and brazed to a single point of connection. Fascia panels, top, middle and bottom cover panels shall be removable for access for easy installation and maintenance of headwall services.
Medical Gas Manifold
Medical gas distribution shall be supplied by pipe drops via the service chase or single point termination zone as indicated on the drawing. All Medical Gas Outlets and piping shall be brazed and tested in accordance with NFPA 99c. All Medical Gas piping shall be Type L copper pipe. Each outlet,
piping and manifold shall pass a 24 hour standing pressure test.
Wire for standard and critical branch power circuits shall be #10 or #12 (as specified) type THHN stranded copper wire, 600 volt, with heat resistant thermoplastic insulation for hot (black) and neutral (white). Grounds shall be #10 type R THHN stranded copper wire (green). All ground conductors shall be installed in conduit.
Grounding and Bonding
All ground conductors shall be installed in conduit. Each power receptacle shall have a ground conductor connected to a grounding screw. A grounding bus will be installed in each device barrier box at rough in locations, and shall be installed as to insure grounding for the complete power system.
Shall be Hospital Grade 15 or 20 amp, 120 or 277 volt, UL listed and marked Hospital Grade. Quantity and type as shown on drawings. Amico shall furnish and install receptacles with wiring required, as shown on submittal shop drawings. Electrical contractor shall insure compatibility of plug on
accessory equipment to be used with these devices: Duplex Receptacles shall be NEMA style 5-15R or 5-20R, color Ivory for use on normal power circuits, and color Red for use on critical branch power circuits.